- Listed: March 16, 2014 2:01 pm
- Expires: This ad has expired
IIJ-II series fully automatic oil tester is designed as per the requirement of IEC156 Testing Method of Insulation Oil Medium Strength, which via proper programming, has got functions of auto boosting, step down, stirring, display and print out. After breakdown of oil medium, HV break off will operate within 10 seconds to enable an auto isolation of control system partly; this effectively safeguards the personnel security and equipment safety. The tester is suited for testing of the strength of insulating oil medium, as well as for testing of withstands voltage of transformer oil.
1) Fully automatic, control by microcomputer, utilization of new I/O terminal.
2) Numbers of times of testing and idling time of stirring are set via digital setting.
3) Test results displayed and printed out automatically.
4) Strong capability of anti-interference.
5) High accuracy of testing and high capacity of anti-jamming.
6) Easy to operate, small size, light weight which operated in the open air very conveniently and convenient for maintenance
7) Can store 99 groups oil testing data which is convenient to reading and print. The testing data can keep 100 years under the condition of power off.
8) System time can go on running under power off.(the system time just for reference can not be the standard of timing)
Technology and parameter data
1. Power supply: AC220V 10%, 50 HZ
2. Voltage output: 60 KV, 80KV, 100KV
3. Capacity: 1.6KVA, 2.0KVA
4. Pressure increase rate : About 2 KVA/S
5. Pressure testing speed: 3%
6. Break down sensitivity : <2KV
7. Wave form distortion :?3%
8. Time of break down: ?10ms
9. Operational environment: Temperature:0°C-40°C, Humidity: the most relative humidity85%
10. Storage environment : Temperature:-20°C-40°C, Humidity: the most relative humidity75%.
11. Operational height: <150m (can be specially designed if the height is over 1500m)
12. Test cell: Glass, volume 300 to 500 ml, with protective cover and stirrer.
13. Electrodes: Stainless steel spherical 36 mm diameter/hemispherical 25 mm radius, 2.5 mm gap as per IEC-156.
14. Nitial stand time of oil sample: 180 seconds
Intermediate stand time of oil sample: 60 seconds
String time of oil sample: 60 seconds
Number of consecutive test: 6
15. Weight for High voltage and control: Total about 50kg
(oil purifier, oil purification, oil filtration, oil recycling, oil treatment, oil regeneration, oil restoration, oil filtering, waste oil disposal, oil reclaiming, waste oil management, energy saving, oil reconditioned, oil reconstituted, oil restituting, oil recovering ,oil filtration, oil filter, petroleum machine)
For additional equipment information and pricing, email us at
Chongqing Top Oil Purifier Co.,Ltd
Contact: Ms Abby
Skype ID: topoilpurifier
Yahoo messenger ID: topoilpurifier
Hotmail ID: topoilpurifier
Gmail talk ID: topoilpurifier
Your inquiry are warm welcomed. We will give a feedback soon.
806 total views, 1 today
You must be logged in to inquire about this ad.
Information about the ad poster
- Listed by: topoilpurifier
- Member Since: March 16, 2014
Other items listed by topoilpurifier
- high vacuum transformer oil purifier series ZYR/ oil regeneration/ waste oil treatment
- Lubricating oil filtration machine series TYA/ oil recovery
- cooking oil treatment machine/ oil purification system
- oil press filtration machine/ oil recycling/ oil separator
- Mobile insulating oil purifier series ZYM
Popular Ads Overall
- 20-14-8001-2120314 Chain mortiser HAFFNER (used) (18018 views)
- 2007 Bobcat S330 (3243 views)
- STRIEBIG Vertical panel saw CONTROL 6224 (2004) FOR SALE (2580 views)
- Hot sale! High quality cnc router machine (2539 views)
- China high quality and cheaper jaw crusher, fine jaw crusher,european type jaw crusher (2467 views)
- GANNOMAT Dowel boring machine ELITE 50 (2003) FOR SALE (2381 views)
- BRANDT Edgebander KD 67C (2001) FOR SALE (2239 views)
- BRANDT Edgebander KD 78CF (1999) FOR SALE (2231 views)
- PANHANS Beamsaw EUROSTAR (2002) FOR SALE (2180 views)
- ALL KINDS OF KOMATSU EQUIPMENT (2100 views)